|Support Center Novaspec Pro|
What is the beam height ?
Beam height is 15mm above the base of the cuvette. Recommended fill level is at least 20mm.
Can I change the bulb myself ?
The tungsten lamp in the Novaspec can be replaced easily by you in your lab.
(80-2115-33 Novaspec Tungsten lamp)
What is single beam technology ?
The Novaspec is a single beam instrument.
Single beam refers to the fact that the light just passes through the sample holder straight to the detector. An initial reference is required to standardise the instrument before analysis can begin. Single beam instruments are generally very simple and economical to purchase.
Product Name Code No. Novaspec Pro Spectrophotometer 80-2120-50
Product Name Code No.
For absolute accuracy in your readings, take a reference measurement (zero reading), just before your sample readings. Regularly store a new default baseline, especially after changing a lamp.
Cuvettes - choose carefully
It is important to choose the best cuvette for your application. We recommend the use of Biochrom semi-micro disposable cuvettes, as some other makes may not fit reproducibly in the cell holders.
The instrument beam height of 15mm should be taken into consideration when choosing low volume cuvettes.
For best results choose quartz cuvettes.
Check your cuvette is filled sufficiently to at least 20mm from the base.
Check your cuvette is free from finger prints and scratches on the faces.